SDM successful at Vinacomin Cua Ong Site
Determination of slurry density in the coal processing plant.
Pipe diameter: 250mm (10 inch)
Pipe material: Steel
Solids: Magnetite Fe3O4
Density: 1.200 – 2.300 g/l
Temperature: 7°C – 40 °C
Vinacomin wanted to prove the performance of the SDM with an indication out of a 30-day trial. This instrument was used to measure the density in the coal flocculation tanks. This trial consisted of two different sampling methods. With each sampling method being examined four times. These samples were both measured manually and with the Rhosonics SDM. Afterwards, the density values were evaluated and determined how much the SDM deviated from the actual results. They have applied the SDM in one of their largest capacity processing plants, which is named Vinacomin Cua Ong Site.
The Rhosonics SDM Slurry Density Meter is a good solution to measure the real-time density in different processes in the mineral processing industry. The measurements are reliable, stable and friendly to the work environment. Since the device is non-nuclear, calibration is made easy and RSO training for radiation safety is not needed.
The first trial that was done, was by adding water each time to the new sample. The starting sample had a density of 1650 g/l. In the first trial, water had been added three times. Therefore, a total of four samples had been taken. Out of these four samples the total average deviation was 0,62%. The graph on the right showcases all the measurement results from the SDM compared to the manual results.
The second trial that was done, was by adding magnetite (Fe3O4) to each new sample. For the first sample, the starting density was 1337 g/l. Like the first trial, four samples had been taken each with a different density. Out of these four samples the total average deviation was 0,725%. The graph on the right showcases all the measurement results from the SDM compared to the manual results.
During a period of two months (from November 2020) the ultrasonic density meter from Rhosonics was evaluated by a team of experts from the Cua Ong Coal Company. The SDM fulfilled the requirements of Vinacomin, and the team was satisfied with the easy calibration and stable measuring results.
“We can confirm that the SDM trial at Vinacomin Cua Ong Site. was successful. Therefore, we will be integrating the Rhosonics SDM in our plant by the end of 2021.”
Pham Hong Thanh, Deputy Director, Vinacomin Cua Ong Site.
You can read the whole story about the success of Vinacomin in the case study. Click on the image below to download the case study.
Want to know more?
- Case Study: SDMs at Agnico Eagle Meadowbank
- Case Study: SDM Density Meters at OCP Beni Amir
- SDM: Slurry Density Meter Brochure (English)
- SDM: Slurry Density Meter Brochure (Spanish)
- SDM: Slurry Density Meter Brochure (Portuguese)
- SDM: Slurry Density Meter Brochure (French)
- SDM: Slurry Density Meter Brochure (Chinese)