Blog: What to do with end-of-life nuclear sources?
Posted on Friday, 6 December 2019 at 14:39
What am I going to do with the old source? This question comes up when radiation sources of density gauges are reaching the end-of-life phase.
The sources of today were purchased about thirty years ago. You can imagine that this was done by different employees. The employee who is qualified as RSO (Radiation Safety Officer) is responsible for maintenance of the sources on-site and has to make a decision what to do with the end-of-life sources of density meters. There are three options, which are discussed in this blog.
Step 1: Disposal options
When the source of the density meter is beyond the usable life, the customer has three options:
1. Disposal to the original manufacturer, 2. Store the source in a bunker, 3.Replace the detector to extend the usable life.
Step 2: Make a selection
Most companies want to get rid of their nuclear density meter gauge because of the health and safety risks, the licenses, trainings and costs. Therefore, more and more companies will choose for the disposal of the source.
Step 3: External company
Some have teams to remove the source, but normally, the manufacturer or supplier of radioactive sources is caring for disposal of the sources. In some countries the manufacturer is obligated to offer disposal services. In other countries, end-users have to arrange the disposal themselves.
You can contact Rhosonics to help with this process.
Step 4: Disposal process
When the external company receives an inquiry, they will send a spreadsheet to fill out. This give them more information about the date of manufacture, the serial number and the gauge in which it is contained. Sometimes they ask for a picture of the device and source tag. With this information, they can verify if it is a source they can take ownership of. Then they require a leak test, which is a process where you wipe the density gauge around and near the opening with a cotton swab. Which is then tested for trace amounts of radioactive material. This is to confirm that the device is not leaking prior to shipment.
Step 5: Returning source to Manufacturer
Depending on the location, they can also provide shipping assistance. Transportation can be an issue if the source manufacturer is located in another country.
Step 6: Selection and price indication
If all is clear, they will agree on payment terms. The associated disposal costs are mostly driven by the isotope and activity. Their costs for disposal of the source are for example about 7.000,- USD in the US. But only if there are no additional costs to solve safety problems with the radiation protection shielding.
Please contact Rhosonics if you want more information about this process.
Do you want to learn more about this topic? Then please read our other blogs as well: